I want to calibrate the gc for hydrogen detection. Any simple laboratory procedure to do this. How accurately i should take the gas in ml.
for eg if i take 2ml of gas with hamilton syrrenge it wont be 2ml in the syrrenge because volume depends on the partial pressure of the vessel from which i have been taking the gas.What about the commercially available calibration gases. Please suggest some suppliers
By sepp/carla on Friday, August 27, 2004 - 01:48 am:
I guess that's the same undergraduate student, who has asked a few days ago.
or buy a book and a gas tight syringe.